|Accuracy||±1% of reading at rates>0.2 mps|
|Response Time||0-999 seconds, user-configurable|
|Rate Units||Meter, Feet, Cubic Meter, Liter, Cubic Feet, USA Gallon, Imperial Gallon, Oil Barrel, USA Liquid Barrel, Imperial Liquid Barrel, Million USA Gallons. Users configurable|
|Totalizer||7-digit totals for net, positive and negative flow respectively|
|Liquid Types||Virtually all liquids|
|Security||Setup values Modification Lockout. Access code needs unlocking|
|Display||4×8 Chinese characters or 4×16 English letters|
|Communication Interface||RS-232C, baud-rate: from 75 to 57600. Protocol made by the manufacturer and compatible with that of the FUJI ultrasonic flow meter. User protocols can be made on enquiry.|
|Transducers||Model M1 for standard, other 3 models for optional|
|Transducer Cord Length||Standard 2×5 meters, optional 2x 10 meters|
|Power Supply||3 AAA Ni-H built-in batteries. When fully recharged it will last over 10 hours of operation.100V-240VAC for the charger|
|Data Logger||Built-in data logger can store over 2000 lines of data|
|Manual Totalizer||7-digit press-key-to-go totalizer for calibration|
|Handset Weight||514g (1.2 lbs) with batteries|
Carbon steel,stainless steel,cast-iron,copper,aluminum,FRP and other
uniform pipeline,liner is allowable.
Adopt high-precision calibration equipment, professional EMC test equipment, regularly sent to the national quality institute for inspection, strict control production process.
1. V-method Installation
V-method installation is the moswidely used mode for daily measurement with pipe inner diameters ranging from 15 millimeter to 200 millimeter. It is also called reflective mode or method.
2. Z-method Installation
Z-method is commonly used when the pipe diameter is between 300 millimeters and 500 millimeters.
3. W-method Installation
W-method is usually used on plastic pipes with a diameter from 15 millimeters to 100 millimeters.